Augmented reality-aided unicompartmental knee arthroplasty
Journal of Experimental Orthopaedics volume 9, Article number: 88 (2022)
To illustrate a surgical technique for augmented reality (AR)-assisted unicompartmental knee arthroplasty (UKA) and report preliminary data.
We developed an AR-based navigation system that enables the surgeon to see the tibial mechanical axis superimposed on the patient’s leg in addition to the tibial cutting angle. We measured the tibial resection angle in 11 UKAs using postoperative radiographs and calculated the absolute difference between preoperative target angle and postoperative measured angle. The target angle was determined for each patient: mean values were 0.7° ± 1.0° varus in coronal alignment and 5.3° ± 1.4° posterior slope in sagittal alignment.
The angles measured on postoperative radiographs were 2.6° ± 1.2° varus in the coronal plane and 4.8° ± 2.5° posterior slope in the sagittal plane. The absolute differences between the target and measured angles were 1.9° ± 1.5° in coronal alignment and 2.6° ± 1.2° in sagittal alignment. No patients experienced complications, including surgical site infection and periprosthetic fracture.
The AR-based portable navigation system may provide passable accuracy in terms of proximal tibial resection during UKA.
Level of Evidence
Unicompartmental knee arthroplasty (UKA) is an attractive option to treat unicompartmental end-stage knee osteoarthritis and osteonecrosis . Accurate tibial bone cutting is one of the most essential determinants for obtaining proper limb alignment after UKA . Malalignment of the lower limb can increase the risk of postoperative complications including contralateral compartmental osteoarthritis, component loosening, and component wear, following UKA .
Computer-assisted navigation systems have contributed to improvement of the accuracy of bone cutting in knee arthroplasty. These navigation systems can be divided into three groups: image-based large-console navigation, imageless large-console navigation, and portable navigation . Both imageless large-console navigation and portable navigation avoid radiation exposure as they do not require preoperative imaging, such as computed tomography . The surgeon can employ a portable navigation system without adding significant extra cost because it does not require expensive equipment .
Augmented reality (AR) technology shows great promise in computer-assisted surgery due to its unique ability to fuse live images and synthetic computer-generated images [4, 8]. We developed a portable navigation system applied to AR technology that projects the tibial mechanical axis and the varus and posterior slope angles of tibial cutting guide on real world through a smartphone display . This is the first study of tibial bone cutting during UKA assisted with AR technology. The aim of this preliminary clinical study was to assess the accuracy of the AR-based portable navigation system for proximal tibial resection during UKA.
This retrospective single-arm cohort study was approved by the institutional ethics board. We reviewed 11 consecutive UKAs performed using the AR-based portable navigation system to proximal tibial resection between January 2020 and November 2021. All patients provided written informed consent.
All surgeries were performed by one surgeon (ST) using a Persona Partial Knee System (Zimmer-Biomet, Warsaw, IN). Tibial bone resection was performed using the AR-based portable navigation system. The target angle of tibial resection was determined for each patient using the original varus angle and posterior slope of the proximal tibia as references. In the coronal plane, we aimed to resect the proximal tibia perpendicular to the mechanical axis of the femur except in patients with preoperative varus angle of the proximal tibia ≥ 6°, in whom we set the target varus angle to 2° or 3° because varus alignment exceeding 4° was reported to be associated with translation in the mediolateral direction . In the sagittal plane, we aimed to resect the proximal tibia with the same angle of the preoperative posterior slope of the medial compartment in each patient. However, we reduced the target angle in patients with a preoperative posterior slope ≥ 6° because excessive posterior slope has been shown to be associated with greater tension of the anterior cruciate ligament and excessive translation in the mediolateral direction . Following tibial bone resection, the distal femur was resected according to the spacer block technique. All prostheses were implanted using Simplex P bone cement (Stryker, Mahwah, NJ).
The tibial resection angle was measured using a standing long-leg radiograph with ImageJ software (US National Institutes of Health, Bethesda, MD). The coronal alignment was measured with reference to the perpendicular line connecting the midpoint of the tibial plateau and the midpoint of the tibial plafond . The sagittal alignment was measured with reference to the perpendicular line connecting the anterior one third of the medial tibial plateau and midpoint of the tibial plafond . Angles were recorded to two decimal places and rounded off to one decimal place. The accuracy of the AR-based navigation system was assessed by calculating the difference between the preoperative target angle and postoperative measured angle. Any complications were recorded with special attention to periprosthetic fracture.
Surgical technique of tibial bone resection using AR-based portable navigation system
AR technology projects digital information onto the real world [11, 12]. The AR-based navigation system enables the surgeon to see the tibial mechanical axis on the patient’s leg and the tibial cutting angle in real-time.
The extramedullary tibial cutting guide of the AR-based navigation system carries two markers with Quick Response (QR) codes (Figs. 1 and 2). The extramedullary tibial cutting guide of the AR-based navigation system was set on the patient’s lower leg in a similar manner to the standard cutting guide. First, the ankle clamp was placed proximal to the malleolar. Second, the surgeon inserted one pin to fix the cutting guide parallel to the anteroposterior axis of the tibia because the AR-based navigation system was programmed to recognize the direction of the pin as the anteroposterior axis (Fig. 2).
The sensor of the AR-based navigation system is the camera of the smartphone. The smartphone camera recognizes the QR code of the extramedullary tibial cutting guide. Using a pointer marked with a QR code, the surgeon registers three bony landmarks: the most prominent point of the medial malleolus, the most prominent point of the lateral malleolus, and the tibial center on the tibial plateau (Fig. 3). Visualization of registration points is a major advantage of the AR-based navigation system, which enables surgeon to easily recognize whether the registered point is wrong or not (Fig. 4). The navigation system creates a 3-dimensional tibial coordinate system to express the position of the tibial cutting guide. The three lines constituting the 3-dimensional coordinate system of AR-based navigation are: (1) the tibial mechanical axis; (2) the tibial anteroposterior axis; and (3) the cross product of these two tibial axes. The tibial mechanical axis is defined as the line connecting the center of the ankle and the tibial center on the tibial plateau. The registration of medial compartments of the tibial plateau is used for the resection level of the tibia, allowing the surgeon to view the proximal tibia resection level.
After completing registration, the AR-based navigation system enables the surgeon to view the reference lines superimposed on the tibia on their smartphone display (Fig. 1). The surgeon can also view the angles of varus/valgus and posterior slope on the display (Fig. 2). The surgeon fixes the tibial resection block while viewing these angles. As the guide pinhole below the tibial prosthesis can be a stress point increasing the risk of tibial stress fracture , the cutting block of the AR-based navigation system has only one pinhole that is located at the cross point of the vertical and horizontal bone cut lines (Fig. 2). Following fixation of the tibial resection block, the surgeon cuts the proximal tibia in the standard manner.
After tibial bone resection, the AR-based navigation system allows the surgeon to verify the actual cutting angle and depth of resection. The display of the smartphone shows varus/valgus angle, tibial slope angle, rotation angle, and the thickness of bone resection (Fig. 5).
The patient characteristics and results of radiographic measurement of the patients are summarized in Table 1.
With regard to coronal alignment, the mean ± standard deviation of postoperative varus angle was 2.6° ± 1.2°. With regard to sagittal alignment, postoperative posterior slope angle was 4.8° ± 2.5°.
The absolute differences between the preoperative target resection angles and postoperative measured angles were 1.9° ± 1.5° in coronal alignment and 2.6° ± 1.2° in sagittal alignment.
No patients experienced complications, including surgical site infection and periprosthetic fracture.
We reported the surgical technique of AR-based portable navigation for tibial bone resection during UKA. AR technology has attracted increasing interest in surgical practice . In orthopedic surgery, AR has been applied to a wide spectrum of procedures, such as total knee arthroplasty (TKA), total hip arthroplasty, tumor resection, arthroscopic surgery, and fracture treatment [2, 5, 6, 8, 11, 12, 14, 18,19,20]. This is the first report applying AR technology to UKA (Table 2).
There are several advantages to using AR-based portable navigation for UKA. First, the AR-based navigation system can visualize the mechanical axis and registration points. Other navigation systems do not have this capability. The visualization may be useful not only for enhancing accuracy of surgical procedures, but also in surgical education for inexperienced surgeons.
Second, the running costs for use of the AR-based portable navigation in UKA are low. The AR-based navigation system can be used with an application downloaded to the surgeon’s own smartphone, several QR code markers, and the dedicated extramedullary guide. In addition, neither an assistant nor additional equipment is required in the nonsterile zone because the operation of the AR-based navigation system is accomplished by the surgeon in the sterile zone alone.
Third, no extra pins are needed to use the AR-based navigation system in UKA because the extramedullary cutting guide carries QR codes for registration. Pinholes located in the proximal tibial plateau are associated with risk of periprosthetic fracture . Therefore, avoiding the insertion of extra pins to attach the navigation sensor may reduce the risk of periprosthetic fracture.
This study had several limitations. First, this was a preliminary study in a single-arm cohort with small sample size. Although a pilot study was crucial to assess the feasibility of a larger comparative study, it must be noted that this study could not conclusively demonstrate the clinical effectiveness of the AR-based navigation system. Second, the results lack generalizability because all surgeries were performed by a single surgeon at a single institute, and surgeon’s experience has been shown to affect the clinical and radiographic outcome of UKA .
The AR-based navigation system may be an effective option to enhance the accuracy of tibial bone resection during UKA. This preliminary study provided important information for future studies to investigate the clinical effectiveness and safety of the AR-based navigation system.
Availability of data and materials
The datasets used and/or analysed during the present study are not publicly available. Data are, however, available from the corresponding author on reasonable request.
Total Knee Arthroplasty
Unicompartmental Knee Arthroplasty
Brumby SA, Carrington R, Zayontz S, Reish T, Scott RD (2003) Tibial plateau stress fracture: a complication of unicompartmental knee arthroplasty using 4 guide pinholes. J Arthroplasty 18:809–812
Chytas D, Malahias MA, Nikolaou VS (2019) Augmented reality in orthopedics: current state and future directions. Front Surg 6:38
Chui EC, Lau LC, Kwok CK et al (2021) Tibial cutting guide (resector) holding pins position and subsequent risks of periprosthetic fracture in unicompartmental knee arthroplasty: a finite element analysis study. J Orthop Surg Res 16:205
Diana M, Marescaux J (2015) Robotic surgery. Br J Surg 102:e15-28
Fucentese SF, Koch PP (2021) A novel augmented reality-based surgical guidance system for total knee arthroplasty. Arch Orthop Trauma Surg 141:2227–2233
Iacono V, Farinelli L, Natali S et al (2021) The use of augmented reality for limb and component alignment in total knee arthroplasty: systematic review of the literature and clinical pilot study. J Exp Orthop 8:52
Jones CW, Jerabek SA (2018) Current role of computer navigation in total knee arthroplasty. J Arthroplasty 33:1989–1993
Laverdière C, Corban J, Khoury J et al (2019) Augmented reality in orthopaedics: a systematic review and a window on future possibilities. Bone Joint J 101:1479–1488
Matziolis G, Mueller T, Layher F, Wagner A (2018) The femoral component alignment resulting from spacer block technique is not worse than after intramedullary guided technique in medial unicompartimental knee arthroplasty. Arch Orthop Trauma Surg 138:865–870
Minoda Y, Hayakawa K, Hagio K, Konishi N, Tamaki T, Iwakiri K (2020) Usefulness of an accelerometer-based portable navigation system for total knee arthroplasty: a multicenter prospective randomized controlled trial. J Bone Joint Surg Am 102:1993–2000
Ogawa H, Hasegawa S, Tsukada S, Matsubara M (2018) A pilot study of augmented reality technology applied to the acetabular cup placement during total hip arthroplasty. J Arthroplasty 33:1833–1837
Ogawa H, Kurosaka K, Sato A, Hirasawa N, Matsubara M, Tsukada S (2020) Does an augmented reality-based portable navigation system improve the accuracy of acetabular component orientation during THA? A randomized controlled trial. Clin Orthop Relat Res 478:935–943
Petterson SC, Blood TD, Plancher KD (2020) Role of alignment in successful clinical outcomes following medial unicompartmental knee arthroplasty: current concepts. J Isakos 5:224–228
Pokhrel S, Alsadoon A, Prasad PWC, Paul M (2019) A novel augmented reality (AR) scheme for knee replacement surgery by considering cutting error accuracy. Int J Med Robot 15:e1958
Price AJ, Alvand A, Troelsen A et al (2018) Knee replacement. Lancet 392:1672–1682
Sekiguchi K, Nakamura S, Kuriyama S et al (2019) Effect of tibial component alignment on knee kinematics and ligament tension in medial unicompartmental knee arthroplasty. Bone Joint Res 8:126–135
Suda Y, Takayama K, Ishida K et al (2020) Improved implant alignment accuracy with an accelerometer-based portable navigation system in medial unicompartmental knee arthroplasty. Knee Surg Sports Traumatol Arthrosc 28:2917–2923
Tsukada S, Ogawa H, Nishino M, Kurosaka K, Hirasawa N (2019) Augmented reality-based navigation system applied to tibial bone resection in total knee arthroplasty. J Exp Orthop 6:44
Tsukada S, Ogawa H, Nishino M, Kurosaka K, Hirasawa N (2021) Augmented reality-assisted femoral bone resection in total knee arthroplasty. JBJS Open Access 6:e21
Tsukada S, Ogawa H, Hirasawa N, Nishino M, Aoyama H, Kurosaka K (2022) Augmented reality- vs accelerometer-based portable navigation system to improve the accuracy of acetabular cup placement during total hip arthroplasty in the lateral decubitus position. J Arthroplasty 37:488–494
Zambianchi F, Digennaro V, Giorgini A et al (2015) Surgeon’s experience influences UKA survivorship: a comparative study between all-poly and metal back designs. Knee Surg Sports Traumatol Arthrosc 23:2074–2080
There was no external source of funding to perform this study.
Ethics approval and consent to participate
The study protocol was approved by the institutional review board of Hokusuikai Kinen Hospital.
Consent for publication
The publication of this study was approved by the institutional review board of Hokusuikai Kinen Hospital.
One of the authors (ST) certifies receipt of personal payments during the study period, in amounts of less than USD 10,000 from Zimmer-Biomet Japan and Stryker Japan.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Tsukada, S., Ogawa, H., Kurosaka, K. et al. Augmented reality-aided unicompartmental knee arthroplasty. J EXP ORTOP 9, 88 (2022). https://doi.org/10.1186/s40634-022-00525-4