- Original Paper
- Open Access
Accuracy, inter- and intrarater reliability, and user-experience of high tibial osteotomy angle measurements for preoperative planning: manual planning PACS versus semi-automatic software programs
Journal of Experimental Orthopaedics volume 9, Article number: 44 (2022)
To compare the accuracy, inter- and intrarater reliability, and user-experience of manual and semi-automatic preoperative leg-alignment measurement planning software for high tibial osteotomy (HTO).
Thirty patients (31 lower limbs) who underwent a medial opening wedge HTO between 2017 and 2019 were retrospectively included. The mechanical lateral distal femur angle (mLDFA), mechanical medial proximal tibial angle (mMPTA), and planned correction angle were measured on preoperative long-leg full weight-bearing radiographs utilising PACS Jivex Review® v5.2 manual and TraumaCad® v2.4 semi-automatic planning software. Independent measurements were performed by four raters. Two raters repeated the measurements. Accuracy in the standard error of measurement (SEM), inter- and intrarater reliability, and user-experience were analysed. Additionally, measurements errors of more than 3° were remeasured and reanalysed.
The SEMs of all measured varus malalignment angles and planned correction angle were within 0.8° of accuracy for both software programs. Measurements utilising the manual software demonstrated moderate interrater intraclass correlation coefficient (ICC)-values for the mLDFA and mMPTA, and an excellent interrater ICC-value for the correction angle (0.810, 0.779, and 0.981, respectively). Measurements utilising the semi-automatic software indicated excellent interrater ICC-values for the mLDFA, mMPTA, and correction angle (0.980, 0.909, and 0.989, respectively). The intrarater reliability varied substantially per angle, presenting excellent intrarater agreements by both raters (ICC > 0.900) for the correction angle in each software program as well as poor-to-excellent ICC-values for the mLDFA (0.282–0.951 and 0.316–0.926) and mMPTA (0.893–0.934 and 0.594–0.941) in both the manual planning and semi-automatic software. Regarding user-experience, semi-automatic software was preferred by two raters, while the other two raters had no distinctive preference. After remeasurement of five outliers, excellent interrater ICC-values were found for the mLDFA (0.913) and mMPTA (0.957).
Semi-automatic software outperforms the manual software when user-experience and outliers are considered. However, both software programs provide similar performance after remeasurement of the human-related erroneous outliers. For clinical practice, both programs can be utilised for HTO planning.
Level of evidence
Diagnostic study, Level III.
High tibial osteotomy (HTO) to correct varus malalignment of the leg is an effective joint-preserving surgical technique to treat medial compartment knee osteoarthritis (OA) in young, active patients [9, 15, 23]. By realigning the mechanical weight-bearing leg axis, the weight-bearing forces are redistributed, decreasing the forces on the medial knee compartment and therefore preventing progression of medial OA [7, 18, 34]. Preoperatively, a malalignment analysis and measurement of the planned correction angle are determined by knee joint angle measurements on long-leg full weight-bearing radiograph . It is important to measure these angles accurately in order to plan and achieve an optimal deformity angle of correction during surgery, since measurement error affects accuracy [20, 21, 28, 33].
Inaccurate preoperative osteotomy planning can be caused by known insufficiencies including inconsistent patient positioning and system setup during radiograph acquisition [2, 3, 5, 22, 33], and lack of awareness of unintentional alteration of adjacent joints [1, 12]. Other than the aforementioned insufficiencies, the accuracy of the preoperative angle measurements and planning of the correction angle utilising software programs is important to take into account.
Various manual and semi-automatic software programs can be utilised to perform the deformity analysis and determine the planned correction angle. Research to date has provided excellent rater agreements for both types of software programs with inter-, and intrarater intraclass correlation coefficients (ICCs) between 0.841 and 0.998 (Table 1) [6, 20, 21, 25,26,27, 33]. Apart from the ICC, there are other possibilities to evaluate accuracy in digital planning, including the measurement error. Because of the huge influence of measurement errors on complication rates and outcome, the measurement error should be considered if accuracy of digital planning will be assessed. Nevertheless, evidence regarding measurement errors for lower limb deformity angles is scarce. To the best of the authors’ knowledge, only Nerhus et al. (2017) reported measurement errors for the femoral and tibial malalignment angles with values up to 2.1° . Measurement errors of 2° imply a substantial risk of unintended over- or underestimation of the size of the varus malalignment, since under- or overcorrection of 1° in the coronal plane can result in progression of medial OA [28, 32]. Additionally, no previous study has investigated the measurement error of the planned correction angle while specifically utilising manual or semi-automatic planning software programs.
Consequently, as current literature does not report all required performance metrics of both types of planning software programs, this study compares the accuracy of manual and semi-automatic software programs in measurement error size, inter- and intrarater reliability, and user-experience for deformity analysis as well as measurements of planned correction angle in HTO procedures. The user-experience of these software programs for analysing deformity and measuring the planned correction angle has not been published previously. The authors hypothesise that the semi-automatic planning software measurements produce lower measurement errors, higher inter- and intrarater reliability, and increased user-experience.
This study was developed in accordance with the Guidelines for Reporting Reliability and Agreement Studies (GRASS) .
Patients who underwent a medial opening wedge HTO between 2017 and 2019 were potentially eligible for inclusion in the present study. Patients who underwent an additional distal femur osteotomy or supramalleolar ankle osteotomy were excluded (Fig. 1) . Thirty-one osteotomies (15 right-knee and 16 left-knee procedures) in 30 patients (26 males and 4 females) with a mean age of 45 (± 12) years, BMI of 26.3 (± 3.3) kg/m2, Kellgren and Lawrence OA grade  of 2 (min. 1 – max. 3), and a mean varus deformity of 5.4° (± 2.3°) were included.
Long-leg full weight-bearing radiographs were acquired preoperatively utilising a Siemens Healthineers’ Thorax/Multix FD SD® (Siemens Healthcare GmbH, Germany) digital radiography system with a universal grid. The X-ray beam was centred at the knee and was positioned at a fixed distance of 300 cm from the patient. Settings of 77–90 kV were utilised, depending on the tissue characteristics and length of the lower extremity. Patients were required to stand, bearing full weight on both legs with 0° of leg extension in the anterior-posterior direction with the patella centralised over the distal femoral condyles. Anterior-posterior and lateral radiographs of the affected knee were acquired with the same X-ray system, utilising a 60 kV setting and a universal grid positioned approximately 120 cm from the source to the detector.
For malalignment analysis, the mechanical lateral distal femur angle (mLDFA) and the mechanical medial proximal tibial angle (mMPTA) were measured. The planned correction angle was measured utilising Miniaci’s method [6, 10, 19].
Manual planning software angle measurements
Basic tools are provided by the manual PACS software Jivex® v5.2 (VISUS Technology Transfer GmbH, Germany), which include circle, ruler, and straight-line tools. An overview of the aforementioned angle measurements is included in Fig. 2. To measure the mLDFA, the centre of the femoral head was determined utilising the electronic option ‘circle with centre-point’. Subsequently, the option ‘open angle measurement’ was selected, and one straight line was drawn from the previously selected femoral centre-point to the centre of the knee; a second straight line was drawn between the two most distal (convex) points of the femoral condyles. For the mMPTA measurement, only the ‘open angle measurement’ was required, so a straight line was drawn between the most convex points of the medial and lateral tibial plateaus, and a second straight line was drawn between the centre of the knee and centre of the talar dome, which represents the mechanical tibial axis. Finally, the correction angle was calculated based on Miniaci’s measurement method , as explained by Elson et al. : first, the position of Fujisawa’s point, based on a weight-bearing axis of 62.5% through the tibial plateau, was calculated utilising the ‘line-relation measurement’ option. Second, the option ‘open angle measurement’ was utilised, and a straight line was drawn, representing the Mikulicz line (preoperative weight-bearing line), as was a straight line from the centre of the hip through Fujisawa’s point to the height of the talar dome. Finally, the option ‘open angle measurement’ was utilised to calculate the correction angle between the two straight lines intersecting at the hinge point. The software program reported the angle measurements to one decimal.
Semi-automatic software angle measurements
TraumaCad v2.4 (VoyantHealth, BrainLab Company, Israel) includes two semi-automatic tools to measure the deformity angles and planned correction angle. The ‘Knee Limb Alignment Analysis tool’ was utilised to measure the mLDFA and mMPTA. In this program, the user selects the centre of the hip based on a three-point circle method: the top of the greater trochanter, the most distal points of the medial and lateral femoral condyles, the most distal points of the medial and lateral tibial plateau, and the most medial and lateral points of the tibial plafond. Then, the ‘High Tibial Osteotomy tool’ was utilised to calculate the HTO correction angle. Figure 3 presents an overview of the semi-automatic software’s joint angle measurements. The software program provides the results in integers.
Three orthopaedic knee surgeons (R1, R2, and R3) and the primary investigator (R4), who was trained by the orthopaedic knee surgeons, independently performed measurements utilising the two software programs. Uniformity of the measurement methods for each radiological parameter was ensured by a single training session prior to commencing the measurements. R1 and R4 performed the measurements again after approximately 6 weeks with a different presentation order of the cases. Because of technical limitations, it was not possible to blind the raters from the subjects’ patient identification numbers.
The user-experience of the software programs was investigated utilising the Post Study System Usability Questionnaire (PSSUQ), which is widely employed to assess user-experience and offers excellent reliability [16, 17]. Its 19 questions must be answered on a 7-point scale, where 1 indicates ‘Strongly Agree’ and 7 indicates ‘Strongly Disagree’. These questions are included in Additional File 1. The PSSUQ’s output includes an average overall score, average System Usefulness (SYSUSE), average Information Quality (INFOQUAL) and an average Interface Quality (INTERQUAL) scores .
The required sample size for a reliability analysis of the continuous angle measurements was determined by the predicted ICC of the interrater reliability using Temel et al.’s method . For the manual and semi-automatic software measurements, ICC-values around 0.800 and 0.900 were predicted, respectively. With a power of 0.8 and a significance level of P < 0.05, a minimum of 31 patients for the manual software and 10 for the semi-automatic software were required.
The accuracy of each angle was determined via the standard error of measurement (SEM), which was derived from the formula SEM = SD * sqrt(1 – ICC) [30, 31], and includes the standard deviation (SD) of the measurement error and the ICC. A measurement error was defined as the difference between the individual rater’s measurement and the overall rater average of the angle measurement on the same long-leg full weight-bearing radiograph.
Interrater reliability analyses were performed utilising a two-way random effects model with the absolute agreement of the average measurements of all raters. A two-way random effects model with the absolute agreement of the single measurements was utilised to determine the intrarater reliability. ICC-values were reported with corresponding 95% confidence intervals (CI). Values between 0–0.500, 0.500–0.750, 0.750–0.900, and 0.900–1.000 indicate poor, moderate, good and excellent reliability, respectively .
Additionally, the similarity between the software programs for each HTO angle was analysed via a paired t-test (normal distribution) or Wilcoxon signed-rank test (non-normal distribution) to determine any software-related significant mean differences. Normality of the data was assessed utilising the Shapiro-Wilk test, considering non-normality at P < 0.05. Statistical analyses were conducted with SPSS version 26.0 (IBM SPSS Statistics, Inc.; Armonk, NY, USA).
An individual rater's measurement that differed more than 3° from the overall rater average on the same long-leg full weight-bearing radiograph was considered an outlier, and was remeasured by the corresponding rater. Outliers were analysed independently and possible human errors were identified. Subsequently, remeasurement of the outliers was performed to either confirm or reject the identified cause (i.e., human error).
Figure 4 presents the absolute measurement errors of all raters of the mLDFA, mMPTA, and correction angle per software program. Accuracy was calculated via the SEM and was found to be within 0.8° for all HTO angles (min. 0.0 – max. 0.8°) for both software programs.
As shown in Fig. 5A, manual software measurements provided moderate interrater reliability for the mLDFA (0.810; 95%-CI 0.668–0.900) and mMPTA (0.779; 95%-CI 0.613–0.883), and excellent interrater reliability for the planned correction angle (0.981; 95%-CI 0.962–0.992). For the semi-automatic software, excellent interrater reliability scores were found for all angles with ICC-values of 0.980 (95%-CI 0.948–0.994, 0.909 (95%-CI 0.756–0.974) and 0.989 (95%-CI 0.971–0.997) for the mLDFA, mMPTA and the correction angle respectively. The intrarater reliability varied per software (Fig. 5B), per rater and per angle, except for the correction angle, which demonstrated excellent agreement (ICC > 0.903) in both software programs. Poor reproducibility of the mLDFA measurements (0.282; 95%-CI -0.065–0.572) was found for R1, whereas R4 had poor reproducibility values for the mMPTA measurements (0.316; 95%-CI -0.030–0.597).
Similarity analyses revealed no software-related significant mean differences for the mLDFA and mMPTA (P = 0.127 and P = 0.782, respectively). There was a significant difference of − 0.5° (P < 0.001) between the programs for the correction angle.
Regarding user-experience, the semi-automatic software was preferred over the manual software by two raters (R3 and R4) based on the SYSUSE and INFOQUAL (Table 2). R1 and R2 had no distinctive preference regarding software programs.
No outliers were identified for the semi-automatic software angle measurements. Five manual software measurements, three mMPTA and two mLDFA measurements, had a measurement error of more than 3° compared to the overall rater average of the angle measurement on the same long-leg full weight-bearing radiograph (Table 3) and were therefore remeasured by R1 and R4. Analyses of the remeasurements of the mMPTA and mLDFA showed SEM values of 0.3° and 0.2°, as well as interrater ICC-values of 0.913 (95%-CI 0.820–0.958) and 0.957 (95%-CI 0.917–0.978), respectively. The intrarater ICC value for the mLDFA by R1 increased to moderate agreement (0.514; 95%-CI 0.210–0.729), while the intrarater ICC value of the mMPTA by R4 increased to excellent (0.934; 95%-CI 0.869–0.968).
The most important finding of the present study is that manual planning has a higher risk of outliers in deformity angle measurements. After remeasurement of the outliers that were related to human errors, both software programs gave similar results in terms of accuracy (< 0.3°) and inter- and intrarater reliability for HTO angle measurements. Regarding user-experience, two out of four raters preferred the semi-automatic software, whereas the other two raters had no distinctive preference. Based on these results, the hypothesis that the performance of semi-automatic software would be superior to manual software programs for HTO planning cannot be rejected nor be confirmed completely.
The accuracy of the preoperative HTO angle measurements is important for optimal deformity correction, which ensures the long-term success of an HTO [20, 21, 28, 33]. However, the study by Nerhus et al. (2017) and the current study are the only reports that quantify the measurement error . Nerhus et al. (2017) report the correlation of repeatability (CR) scores for the mLDFA and mMPTA utilising semi-automatic medical planning software (mediCAD v2.24 osteotomy module); these scores can be converted to acquire the SEM (CR = SEM*2.77). The present study reveals lower measurement errors than Nerhus et al. (2017) for the mLDFA (0.4° vs. 0.7°). For the mMPTA measurements, a lower error was found after remeasurement of the outliers which were related to typographical errors (0.2° vs. 0.7°) . These small differences in accuracy (less than 1°) may not be clinically relevant, since under- or overcorrection of 1° or more has been related to the progression of OA [28, 32]. This insinuates that, in clinical practice, both manual and semi-automatic software programs provide accurate preoperative HTO planning for varus malalignment. Nonetheless, it should be noted that several outliers were identified for the manual software angle measurements.
Regarding interrater reliability, the results herein correspond with the interrater reliabilities of manual and semi-automatic software programs reported in other studies (Table 1). However, the authors found moderate interrater ICC-values for the mLDFA as measured with the manual software. In-depth analysis of the outliers revealed that the moderate interrater ICC-values for the mLDFA that were obtained utilising the manual software seemed to be related to the three incorrect mLDFA measurements by R1. These incorrect measurements probably occurred due to the colour blindness of R1 as the level of colour contrast between the measurement lines and the long-leg full weight-bearing radiograph in the user interface of the manual software program is low. A post-hoc analysis excluding the data of R1, demonstrated a remarkable change in mLDFA measurements utilising the manual software from a mean ICC-value of 0.810 to a mean ICC-value of 0.918. However, this difference is most likely caused by absence of the outliers which were measured by R1, as the remeasured outlier data including the data of R1 shows a similar ICC-value of 0.913.
The reported intrarater reliabilities in this study were moderate for the mLDFA and mMPTA measurements that were utilised with the semi-automatic software, whereas previous literature reveals good-to-excellent intrarater reliability scores (Table 1) [21, 25, 27, 33]. The lower intrarater reliability scores may be related to the smaller sample size measured for the semi-automatic software measurements, and the precision of the semi-automatic software measurements (that were rounded up to integers instead of decimals).
The user-experience was scored utilising the PSSUQ [16, 17], which scores the overall functionality based on the software’s usefulness in its interface, system usability, and information quality. Two raters preferred the semiautomatic software, predominantly because of the SYSUSE. These raters were younger and less experienced (1 and 5 years) with the manual HTO angle measurements compared to the other raters (> 10 years). In spite of the lack of experience with TraumaCad's semi-automatic HTO angles measurements by the raters, these values indicate that the program was user-friendly and useful after only a few HTO angle measurements (N = 10).
The outliers of R1 might have been caused by the potential influence of colour-blindness. The outliers of R4 were most likely related to misinterpretation of the measured angle. In case the outliers remained unnoticed, these could potentially have led to a false indication of the size and the origin of the deformity. One of the three mLDFA outliers was higher than 90°, indicating the deformity would have been located in the femur and may have led to a double osteotomy. With regard to the mMPTA measurements, the lines were made correctly, but the right interpretation of the measured angle was not made. Notwithstanding, the correction angles did not experience poor to moderate reliability and reproducibility measurements, but had excellent reliability and producibility, suggesting the orthopaedic surgeons would have noticed these incorrect measurements during the simulation of the correction.
This study has several strengths and limitations. The main strength is that it is one of the first studies in which accuracy measurements were conducted and the user-experience of manual and semi-automatic software programs was investigated regarding HTO planning. However, this study was limited by software-related restrictions. First, it was not possible to anonymise patient data, which may have allowed raters to recall their doctor-patient interactions. Second, the low-contrast ratio of the manual software program’s user interface potentially resulted in misinterpreted angle measurements by the rater with colour blindness (R1). Furthermore, there was a significant difference in similarity for the correction angle between the software programs. Nonetheless, this difference was lower than 1° and may therefore be clinically irrelevant [28, 32]. Additionally, the results of the semi-automatic software could potentially have been biased by the lower number of measurements compared to the manual software, as well as by a possible learning curve since not all raters had used the semi-automatic software before. These biases may have led to an underestimation of the semi-automatic performance.
In addition to the PSSUQ items to score user-experience, it would be interesting to include the measurement time for all HTO angles. Semi-automatic tools are potentially less labour-intensive compared to manual angle measurements. However, TraumaCad had not been utilised prior to this study, hence the measurement time may have been biased by a learning curve.
For optimal osteotomy planning using long-leg full weight-bearing radiographs, guidelines for system setup and an easy-to-implement protocol for patient positioning should be developed to minimize measurement errors during acquisition of radiographs [2, 3, 5, 22, 33]. Additionally, orthopaedic surgeons have to bear in mind that the alignment of adjacent joints may be unintentionally altered as a result of a planned osteotomy which could lead to undesirable effects, such as secondary OA of the ankle [1, 12]. These unforeseen changes in the alignment of adjacent joints may be limited by using a semi-automatic software program which automatically displays all frontal lower limb angles during simulation of the correction, simultaneously shows (un)wanted angle changes and provides recommendations for alterations [4, 8].
This study investigated the accuracy, inter- and intrarater reliability, and user-experience of HTO angle measurements acquired utilising manual and semi-automatic software. The semi-automatic software outperforms the manual software when user-experience and measurement outliers are considered. However, after remeasurement of the human-related erroneous outliers, no significant differences were found between the software programs in accuracy, or inter- and intrarater reliability, indicating a low risk of unintended under- or overcorrection of the planned varus malalignment correction. Both programs can be utilised for HTO malalignment analysis in clinical setting.
Guidelines for Reporting Reliability and Agreement Studies
High tibial osteotomy
Intraclass correlation coefficients
Mechanical lateral distal femur angle
Mechanical medial proximal tibial angle
Post Study System Usability Questionnaire
Standard errors of measurements
Ahmad SS, Kerber V, Konrads C, Ateschrang A, Hirschmann MT, Stöckle U et al (2021) The ischiofemoral space of the hip is influenced by the frontal knee alignment. Knee Surg Sports Traumatol Arthrosc 29(8):2446–2452
Ahrend M-D, Baumgartner H, Ihle C, Histing T, Schröter S, Finger F (2021) Influence of axial limb rotation on radiographic lower limb alignment: a systematic review. Arch Orthop Trauma Surg. https://doi.org/10.1007/s00402-021-04163-w
Ahrend M-D, Rühle M, Springer F, Baumgartner H (2021) Distance from the magnification device contributes to differences in lower leg length measured in patients with TSF correction. Arch Orthop Trauma Surg. https://doi.org/10.1007/s00402-021-03831-1
Awang N, Sulaiman R, Shapi’i A, AHA R, MFM A, Osman S (2015) A comparative study of computer aided system preoperative planning for high Tibial osteotomy. In: Robinson P, Smeaton AF, Shih TK, Velastin S, Jaafar A, Mohamad Ali N (eds) Badioze Zaman H. Adv Vis Inform Springer International Publishing, Switzerland, pp 189–198
Chua CXK, Tan SHS, Lim AKS, Hui JH (2021) Accuracy of biplanar linear radiography versus conventional radiographs when used for lower limb and implant measurements. Arch Orthop Trauma Surg. https://doi.org/10.1007/s00402-020-03700-3
Elson DW, Petheram TG, Dawson MJ (2015) High reliability in digital planning of medial opening wedge high tibial osteotomy, using Miniaci’s method. Knee Surg Sports Traumatol Arthrosc 23(7):2041–2048
Gao L, Madry H, Chugaev DV, Denti M, Frolov A, Burtsev M et al (2017) Advances in modern osteotomies around the knee: report on the association of sports traumatology, arthroscopy, orthopaedic surgery, rehabilitation (ASTAOR) Moscow international osteotomy congress 2017. J Exp Orthop 6(1):9
Hankemeier S, Gosling T, Richter M, Hufner T, Hochhausen C, Krettek C (2006) Computer-assisted analysis of lower limb geometry: higher intraobserver reliability compared to conventional method. Comput Aided Surg 11(2):81–86
Hunter DJ, Bierma-Zeinstra S (2019) Osteoarthritis. Lancet 393(10182):1745–1759
Jiang X, Xie K, Han X, Ai S, Wu H, Wang L et al (2022) HKA angle—a reliable planning parameter for high tibial osteotomy: a theoretical analysis using standing whole-leg radiographs. J Knee Surg 1(35):54–60
Kellgren JH, Lawrence JS (1957) Radiological assessment of osteo-arthrosis. Ann Rheum Dis 16(4):494–502
Konrads C, Eis A, Ahmad SS, Stöckle U, Döbele S (2021) Osteotomies around the knee lead to corresponding frontal realignment of the ankle. Eur J Orthop Surg Traumatol. https://doi.org/10.1007/s00590-021-03016-x
Koo TK, Li MY (2016) A guideline of selecting and reporting intraclass correlation coefficients for reliability research. J Chiropr Med 15(2):155–163
Kottner J, Audigé L, Brorson S, Donner A, Gajewski BJ, Hróbjartsson A et al (2011) Guidelines for reporting reliability and agreement studies (GRRAS) were proposed. J Clin Epidemiol 64(1):96–106
Lee DC, Byun SJ (2012) High tibial osteotomy. Knee Surg Relat Res 24(2):61–69
Lewis JR (1995) IBM computer usability satisfaction questionnaires: psychometric evaluation and instructions for use. Int J Hum-Comput Interact 7(1):57–78
Lewis JR (2002) Psychometric evaluation of the PSSUQ using data from five years of usability studies. Int J Hum-Comput Interact 14(3–4):463–488
Loia M, Vanni S, Rosso F, Bonasia D, Bruzzone M, Dettoni F et al (2016) High tibial osteotomy in varus knees: indications and limits. Joints 4(2):98–110
Miniaci A, Ballmer FT, Ballmer PM, Jakob RP (1989) Proximal tibial osteotomy. A new fixation device. Clin Orthop 246:250–259
Munier M, Donnez M, Ollivier M, Flecher X, Chabrand P, Argenson J-N et al (2017) Can three-dimensional patient-specific cutting guides be used to achieve optimal correction for high tibial osteotomy? Pilot study. Orthop Traumatol Surg Res 103(2):245–250
Nerhus TK, Ekeland A, Solberg G, Sivertsen EA, Madsen JE, Heir S (2017) Radiological outcomes in a randomized trial comparing opening wedge and closing wedge techniques of high tibial osteotomy. Knee Surg Sports Traumatol Arthrosc 25(3):910–917
Nguyen HC, Gielis WP, van Egmond N, Weinans H, Slump CH, Sakkers RJB et al (2021) The need for a standardized whole leg radiograph guideline: the effects of knee flexion, leg rotation, and X-ray beam height. J Cartil Jt Preserv 1(3):100022
van Outeren MV, Waarsing JH, Brouwer RW, Verhaar JAN, Reijman M, Bierma-Zeinstra SMA (2017) Is a high tibial osteotomy (HTO) superior to non-surgical treatment in patients with varus malaligned medial knee osteoarthritis (OA)? A propensity matched study using 2 randomized controlled trial (RCT) datasets. Osteoarthr Cartil 25(12):1988–1993
Paley D, Herzenberg JE, Tetsworth K, McKie J, Bhave A (1994) Deformity planning for frontal and sagittal plane corrective osteotomies. Orthop Clin North Am 25(3):425–465
Schröter S, Ihle C, Mueller J, Lobenhoffer P, Stöckle U, van Heerwaarden R (2013) Digital planning of high tibial osteotomy. Interrater reliability by using two different software. Knee Surg Sports Traumatol Arthrosc 21(1):189–196
Segev E, Hemo Y, Wientroub S, Ovadia D, Fishkin M, Steinberg DM et al (2010) Intra- and interobserver reliability analysis of digital radiographic measurements for pediatric orthopedic parameters using a novel PACS integrated computer software program. J Child Orthop 4(4):331–341
Sled EA, Sheehy LM, Felson DT, Costigan PA, Lam M, Cooke TDV (2011) Reliability of lower limb alignment measures using an established landmark-based method with a customized computer software program. Rheumatol Int 31(1):71–77
Song SJ, Bae DK (2016) Computer-assisted navigation in high tibial osteotomy. Clin Orthop Surg 8(4):349–357
Temel G, Erdogan S (2017) Determining the sample size in agreement studies. Marmara Med J 30(2):101–112
Tighe J, McManus I, Dewhurst NG, Chis L, Mucklow J (2010) The standard error of measurement is a more appropriate measure of quality for postgraduate medical assessments than is reliability: an analysis of MRCP (UK) examinations. BMC Med Educ 10(1):40
de Vet HCW, Terwee CB, Knol DL, Bouter LM (2006) When to use agreement versus reliability measures. J Clin Epidemiol 59(10):1033–1039
Yang JC-S, Chen C-F, Luo C-A, Chang M-C, Lee OK, Huang Y et al (2018) Clinical experience using a 3D-printed patient-specific instrument for medial opening wedge high tibial osteotomy. Biomed Res Int 2018:1–9
Yazdanpanah O, Karimi Mobarakeh M, Nakhaei M, Baneshi MR (2017) Comparison of double and single leg weight-bearing radiography in determining knee alignment. Arch Bone Jt Surg 5(3):174–180
Zheng K (2014) The effect of high tibial osteotomy correction angle on cartilage and meniscus loading using finite element analysis. University of Sydney
No funding was received.
Ethics approval and consent to participate
Ethical approval was not required because of the retrospective nature of the study.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Laven, I.E.W.G., Schröder, F.F., de Graaff, F. et al. Accuracy, inter- and intrarater reliability, and user-experience of high tibial osteotomy angle measurements for preoperative planning: manual planning PACS versus semi-automatic software programs. J EXP ORTOP 9, 44 (2022). https://doi.org/10.1186/s40634-022-00475-x
- High tibial osteotomy
- Deformity analysis
- Radiographic software
- Preoperative planning, accuracy
- Measurement error